Tapping on the (i) symbol on the sensor channels display will show detailed information about the connected UMB devices, e.g. device software and hardware version information etc.
Note: The MARWIS app supports any kind of UMB device, and any number of UMB devices connected via bluetooth, so this screen is designed to show more than one device, even though you will usually connect to only one MARWIS device.
Tapping on a device in the list opens the “Device Configuration” dialog (if the Bluetooth connection is established).
Here some device settings can be edited.
Note that any configuration changes will be transmitted to the device when the dialog is closed (by returning to the device information screen). Note that in order for the configuration changes to take effect the device must be reset/restarted.
The following parameters can be set:
On the right hand side, the adjustment profiles of the device are shown and can be selected or adjusted.
Note that only valid profiles can be selected. Invalid profiles need to be adjusted to become valid. Invalid profiles are shown in gray.
Note that not all profiles can be adjusted – there might be factory presets that cannot be re-adjusted/overwritten.
Tapping on “adj.” will open the “Adjustment/Alignment” dialog (see below).
An offset for the pyrometer temperature measurement can be set. The actual measured temperature is initially shown.
The available buttons have the following functions:
The app is defined as dashboard application. This means, it displays the measurement values and other sensor information. Even if the app runs in the background (e.g. after pressing the device’s “home” button or switching off the display) it still collects data and transfers them to the server.
In order to really switch off the app and stop the data processing close it as follows:
Double-tapping the home button will show the running/active apps on the iPad/iPhone.
Swiping an app out of the screen to the top will terminate the app.